Installation and commissioning steps of near infrared moisture meter
1. After unpacking, check whether the accessories are complete.
2. Install and tighten the holder for the fixed electrolytic cell (different types of holders) on the host. When the count reaches the end point, stop counting, the end point light is on, and the buzzer sounds, then the electrolyte equilibrium point is found, and normal sample measurement can be performed.
3. Take the electrolytic cell out of the calcium plastic box carefully, take out 2 electrodes, the drying tube (embedded in the calcium plastic box), the injection cock, and the electrolytic cell bottle stopper, and apply a little vacuum grease to all the grinding ports Or Vaseline for use.
4. The silicone rubber pad in the middle of the injection cock has been installed. It is difficult to pierce the injection needle for the first time. You can use a hard needle to prick a few times first, or loosen the injection cock. Place the measuring electrode (thin electrode with 2 small platinum studs), drying tube, bottle stopper, and injection stopcock in the corresponding positions of the cell bottle. Place the installed electrolytic cell on the host, and the whole operation process is completely free of water touch.
5. Turn the electrolyte bottle open and pour the electrolyte above the lower scale line of the cell bottle. Then pour the electrolyte into the electrolytic electrode (with a net) about 2 cm high, place it in the position of the cell bottle, put the white mixing rod into the electrolytic cell liquid. For the electrolyte to be easily stored, the fresh electrolyte is black and color The deeper the electrolyte, the fresher it is. The newly balanced electrolyte, the pool bottle is yellow, the inside of the electrolytic electrode is dark, and naturally turns to light yellow after a few hours. It does not matter if the liquid level of the electrolytic electrode and the cell bottle are not the same, and it will become the same height after a few hours of normal use.
6. Plug the electrolytic electrode (with net) into the electrolytic socket of the host, and plug the measuring electrode plug into the measuring socket of the host. After the electrolytic cell is placed, the left side is inserted into the left side, and the right side is inserted into the right side.
7. Turn on the power switch behind the host, turn on the electrolysis switch on the panel of the host, and turn on the mixing switch. Adjust the mixing speed adjustment knob of the main machine so that the liquid surface has a small vortex shape. Don't be too fast, too fast mixing will beat the electrode.
下一条: No Information
Related Industry Knowledge
- The purchase of a moisture meter must be determined according to your own needs
- The main points of daily maintenance of fast moisture analyzer
- What are the uncertain contents and effects of the feed moisture meter in the measurement?
- Causes of poor repeatability of grain moisture meter
- Analysis of development and influencing factors of rapid moisture analyzer
- About the working principle of feed moisture meter
- Characteristics of feed moisture meter equipment
- How to use food moisture meter
- Analysis of the technical characteristics of near infrared on-line moisture analyzer