The main points of daily maintenance of fast moisture analyzer
Moisture meter is an instrument that can detect the moisture content of various organic and inorganic solid, liquid, gas and other samples, and fast moisture meter is a new type of rapid moisture detection instrument, sub-detection instrument.
Its ring-shaped halogen lamp ensures uniform heating of the sample, simple operation and accurate measurement. While measuring the weight of the sample, the rapid moisture analyzer uses the halogen heating method of the ring tube to quickly dry the sample. During the drying process, the moisture meter continuously measures and immediately displays the moisture content of the sample. After the drying process is completed, the final moisture content value is locked and displayed.
Pay attention to the following items during the maintenance of the rapid moisture meter:
1. When the rapid moisture meter is in a standby state, it can be used without preheating.
2. The minimum sample size is greater than 0.5 grams.
3. The cover of the quick moisture meter should be lifted and released gently.
4. The quick moisture meter should clean the sample tray before placing the sample, and the sample tray holder should be used whenever possible to prevent burns from the hot sample tray.
5. In the process of using the quick moisture meter, you should use your fingers to lightly press the buttons, avoid using nails, and prohibit the use of sharp hard objects.
6. Before using the rapid halogen moisture analyzer, carefully check whether the power supply voltage meets the requirements of the halogen moisture analyzer, whether the socket and the device plug are intact, and do not use it if it is damaged.
7. The last appearance of the test will be saved on the screen until the tare button is pressed. The second subsequent test should be as soon as possible to avoid the loss of water.
In addition, please turn off the main power switch before leaving work or when the instrument is not used for a long time. The power plug without the main switch should be placed around the instrument.
Related Industry Knowledge
- Installation and commissioning steps of near infrared moisture meter
- The purchase of a moisture meter must be determined according to your own needs
- What are the uncertain contents and effects of the feed moisture meter in the measurement?
- Causes of poor repeatability of grain moisture meter
- Analysis of development and influencing factors of rapid moisture analyzer
- About the working principle of feed moisture meter
- Characteristics of feed moisture meter equipment
- How to use food moisture meter
- Analysis of the technical characteristics of near infrared on-line moisture analyzer